The DV 3075 measured value sensor serves as a measurement transducer for Divatec DFM 165 to DFM 200 flow meters and is mounted on the dovetail guide.
The output signal from 4 to 20 mA is proportional to the height of the magnetic float.
The measured value sensor DV 3075 supplies a 4 to 20 mA signal generated by the height of the float in the flow meter (DFM).
The unit consists of a Reed chain, the corresponding valuation and transduction.
The floats of the flow meters have a permanent magnet.
The flow of the medium causes the float to pass through the Reed chain.
The contacts of the Reed chain are operated depending on the height of the float.
The measured values are output as a current signal from 4 to 20 mA. This signal can be processed directly.
The control is carried out through the magnet on the float of the Divatec flowmeter.
Vertical mounting, the Divatec logo facing up.
The equipment is connected with a usual cable without shielding. If electromagnetic interpolations are expected above the verification values of EN 61326 for industrial areas, an armored cable should be used.
• Analog output 4 … 20 mA
• Input 8 … 28 V DC
• Individual, programming adapted to DFM
• Calibration of 11 positions
• Non-volatile storage of values
• Pushbutton 0 to compensate for magnetic environmental influences
• Factory adjustable inertia (low-pass filter) 0.1 … 2.5 s according to customer’s specification
• Accuracy of measurement better than 0.5% measurement accuracy less than 5%
The DV 3000 supplies a standard industry signal of 4 to 20 mA that is generated by an accurate magnetic angle recording of the magnetic float in DFM. With the DV 3000, existing flowmeters (with magnetic floats) can be easily retrofitted to electrically signal the flow and, for example, to be processed in a PLC in order to execute a process control or to represent the flows directly on a screen. Divatec programs this device controlled by a microprocessor for the corresponding DFM.
This guarantees an accurate measurement and evaluation of the flow.
Before you know in which DFM the DV 3000 is mounted.
• The denomination of the fluid is in the corresponding scale. If not stated otherwise, the percentage scale is based on H2O, 20 ° C.
Distribución de Válvulas Técnicas, S.L.
Rambla Josep Maria Jujol, 33 – 35, local E1- F1
08970 – Sant Joan Despí (Barcelona) ·
Teléfono 93 372 23 54 email@example.com
Nuestra web utiliza cookies propias y de terceros para analizar sus hábitos de navegación y elaborar estadísticas sobre su interacción con nuestro sitio web, así como mostrar botones de compartición de contenido en redes sociales. Puede obtener más información en nuestra Política de Cookies